get started using the tester. For additional information, see the MicroScanner2 Cable Verifier. Users Manual on the Fluke Networks website. Safety. Warning. MicroScanner2 Cable Verifier. Accessing the Users Manual. This guide provides basic information to help you get started using the tester. For additional. View and Download Fluke MicroScanner 2 user manual online. Cable Verifier. MicroScanner 2 Test Equipment pdf manual download.
|Published (Last):||9 May 2013|
|PDF File Size:||14.30 Mb|
|ePub File Size:||13.54 Mb|
|Price:||Free* [*Free Regsitration Required]|
MicroScanner2 Getting Started Guide, Rev. 1 | Fluke Networks
Using The Toner Refer to Figures 24 and Page 33 – Connecting to a Telephone Network Wired Fiber Inspection and Cleaning. Welcome to Fluke Networks. The higher the number, the stronger the signal.
Results Screens for Individual Wire Pairs Fluke Networks does not warrant that software will be error free or operate without microscsnner2. Using the Toner in IntelliTone Mode. Fluke Networks warrants that software will operate substantially in accordance with its functional specifications for 90 days and that it has been properly recorded on non-defective media.
For cabling not terminated with an F-connector, use an adapter or hybrid patch cord to connect to the cabling. Connect the tester to the cable.
FLUKE MICROSCANNER 2 USER MANUAL Pdf Download.
In a split pair, continuity from end to end is correct, but is made with wires from different pairs. Navigates through screens and changes settings. The shorted Figure 8 shows that wires 3 and 4 are crossed. SmartTone works with the IntelliTone probe and with analog probes. Punch Down Tools and Termination Tools. Regulatory Information MicroScanner Cable Verifier Users Manual Regulatory Information This equipment generates, uses, and can radiate radio frequency energy, and, if not installed and used in accordance with the manual, may cause interference to radio communications.
Don’t have an account? Telephone cables wired microscaanner2 a star topology Figure 16 are connected together at a bridge tap at the distribution If you connect the tester to the bridge tap, the tester center. Without this termination, the tester cannot detect crossed wires or crossed pairs.
Toning in IntelliTone™ Mode – MicroScanner2
microscanner2 Integrity Networks Alaska Branch. Diagnosing Wiremap Faults Appendix A: Note Use the SmartTone function only on dry pairs of wires that are unterminated at both ends.
Open on Twisted Pair Cabling Figure 6 shows an open on wire 4. Use the probe to find the general location of the tone at a cable rack, patch panel, or behind a wall, as shown in Figure 2.
Fluke MicroScanner 2 User Manual
Page 34 – Connecting to Telephone Networks Wired i Detects the advertised speed of All product names are trademarks of their respective companies. Versiv Cabling Certification System. The the pin wires flash to indicate the fault.
Page 60 MicroScanner Cable Verifier Mocroscanner2 Manual split pair, 15 telephone voltages, 16 —U— universal adapter, 10 —V— version information, 41 voltage detection, 5, 16 —W— wiremap adapter, Copper Cable Testers and Tone Generators. The cable length is Figure 19 shows a good coaxial cable On the individual results screens, shorts are shown only when they are between wires in a pair.