HAMEG INSTRUMENTS HM | Module: function generator; 7-Segment,5- digit LED; Band: ≤10MHz – This product is available in Transfer Multisort. F requency Range 50 mHz 10 MHz,. Output Voltage up to 10 Vpp (into 50 Ω). ☑ Waveforms: Sine Wave, Triangle, Square Wave, Pulse, DC. ☑ D istortion Factor. Funktionsgenerator HM Equipment HAMEG GmbH, Frankfurt, build ?, 4 pictures, 4 schematics, Germany, tubes, semiconductors, Service- or Lab.
|Published (Last):||10 May 2008|
|PDF File Size:||12.20 Mb|
|ePub File Size:||3.48 Mb|
|Price:||Free* [*Free Regsitration Required]|
You can write in English language into the forum not only in Hungarian! Connect the oscilloscope to output.
Funktionsgenerator HM Equipment HAMEG GmbH, Frankfurt
Including the amplitude control the max. Any zeroes preceding the decimal point are dropped. In high signal volt- age ranges, it should be noted that the used terminating resistor must dissipate the corresponding effective power. Stability of the output amplitude Setting: We regret that transportation damage due to poor packaging is not covered by this warranty. Remove both screws on rear panel and, holding case firmly in place, pull chassis forward out of case.
You have chosen to save the following item to a parts list:. Trigger output In the sine, square and triangle modes, the trigger out- put 0 supplies a square signal in synchronism with the output signal. Instead, contact your nearest service center!
Alternatively adjust 0 and 0, until the minimum value is obtained. Accord- ingly, the overdrive indicators are light up in case of posi- tive or negative overdriving.
Sending feedback, please wait Intentional interruption is prohibited. Adjust , until the negative pulse length equals the positive pulse length on the screen.
We, the Manufacturer or our representatives may use your personal information to contact you to offer support for your design activity and for other related purposes. Use a 50 ohms through-termination. In case of cleaning the module the red display pane should not come in contact with alcohol or methylated spirit. General Information Operating condKions: Also some files are djvu so you need djvu viewer to open them. F – Distortion adjustment Setting: Setting range from x0. Frequency variation by FM input Setting: The frequency changes by a factor of Adjust 0 for minimum overshoot or rounding, respec- tively, of the rising edge.
First-time operation of the module Provided that all hints given in the operating instructions of the HM Mainframe were followed – especially for the selection of the correct mains voltage – start of operation consists practically of inserting the module into the right or left opening of the mainframe.
All outputs are short-circuit-proof. HM function generator module,5MHz. By clicking the accept button below, you agree to the following terms.
Providing the finest test equipment solutions since 1992
Adjustment range of the offset voltage Setting: DC offset If the offset knob is ha,eg in its calibrated position, a DC voltage can be superimposed on the output signal. Additional quality fea- tures include the relatively low distortion factor of the generated signals and constant amplitude flatness throughout the entire frequency range of the instrument.
H 68, D mm Weight: If the output of the HM unit comes into contact with components of the circuit under test, which are carrying DC voltage i. Good luck to the repair! In addition, some files are archived, so you need WinZip or WinRar to open that files.
HM Function Generator Module | Overview | Rohde & Schwarz
Szeretnem tudni milyen vegyszerrel lehetne tokeletesen kitakaritani. Set oscilloscope to DC coupling. Set the trigger edge con- trol slope of the oscilloscope to negative edge trigger- ing.
An offset voltage adjusted at the 50 ohms output has no influence upon the trigger signal.